Full text loading...
No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
Performance evaluation of computed radiography systems
1.R. Schaetzing, B. R. Whiting, A. R. Lubinsky, and J. F. Owen, “Digital radiography using storage phosphors,” in Digital Imaging in Diagnostic Radiology, edited by J. D. Newall and C. A. Kelsey (Churchill Living Stone, 1990), pp. 107–138.
2.M. Sonoda, M. Takano, J. Miyahara, and H. Kato, “Computed radiography utilizing scanning laser stimulated luminescence,” Radiology 148, 833–838 (1983).
3.J. A. Seibert, “Photostimulable phosphor system acceptance testing,” in Specification, Acceptance Testing and Quality Control of Diagnostic X-ray Imaging Equipment, edited by J. A. Seibert, G. T. Barnes, and R. G. Gould (AIP, New York, 1994), pp. 771–800.
4.C. E. Willis, R. G. Leckie, J. Carter, M. P. Williamson, S. D. Scotti, and G. Norton, “Objective measures of quality assurance in a computed radiography-based radiology department,” SPIE Med. Imaging 2432, 588–599 (1995).
5.J. A. Seibert et al., “Acceptance testing and quality control of photostimulable phosphor imaging systems,” Report of the American Association of Physicists in Medicine (AAPM) Task Group No. 10 (unpublished, in the final review process).
6.A. R. Cowen, A. Workman, and J. S. Price, “Physical aspects of photostimulable phosphor computed radiography,” Br. J. Radiol. 66, 332–345 (1993).
7.E. Samei, M. J. Flynn, and D. A. Reimann, “A method for measuring the presampled MTF of digital radiographic systems using an edge test device,” Med. Phys. 25, 102–113 (1998).
8.E. Samei and M. J. Flynn, “Physical measures of image quality in photostimulable phosphor radiographic systems,” SPIE Med. Imaging 3032, 338 (1997).
9.J. T. Dobbins III, D. L. Ergun, L. Rutz, D. A. hinshaw, H. Blume, and D. C. Clark, “DQE(f) of four generations of computed radiography acquisition devices,” Med. Phys. 22, 1581–1593 (1995).
10.E. Samei, D. J. Peck, P. L. Rauch, E. Mah, and M. J. Flynn, “Exposure calibration of computed radiography imaging systems (abstract),” Med. Phys. 25, A155 (1995).
11.C. D. Bradford, W. W. Peppler, and J. T. Dobbins III, “Performance characteristics of a Kodak computed radiography system,” Med. Phys. 26, 27–37 (1999).
12.W. Hillen, U. Schiebel, and T. Zaengel, “Imaging performance of a digital storage phosphor system,” Med. Phys. 14, 744–751 (1987).
13.C. E. Floyd, H. G. Chotas, J. T. Dobbins III, and C. E. Ravin, “Quantitative radiographic imaging using a photostimulable phosphor system,” Med. Phys. 17, 454–459 (1990).
14.M. J. Flynn and E. Samei, “Experimental comparison of noise and resolution for 2k and 4k storage phosphor radiography systems,” Med. Phys. 26, 1612–1623 (1999).
15.C. E. Willis, J. C. Weiser, R. G. Leckie, J. Romlein, and G. Norton, “Optimization and quality control of computed radiography,” SPIE Med. Imaging 2164, 178–185 (1994).
16.D. M. Tucker and P. S. Rezentes, “The relationship between pixel value and beam quality in photostimulable phosphor imaging,” Med. Phys. 24, 887–893 (1997).
17.M. Freedman, E. Pe, S. K. Mun, S. C. B. La, and M. Nelson, “The potential for unnecessary patient exposure from the use of storage phosphor imaging systems,” SPIE Med. Imaging 1897, 472–479 (1993).
18.C. E. Willis, “Computed Radiography: QA/QC,” in Practical Digital Imaging and PACS, Medical Physics Monograph No. 28 (Medical Physics Publishing, Madison, 1999), pp 157–175.
Article metrics loading...