Full text loading...
Solid-state fluoroscopic imager for high-resolution angiography: Physical characteristics of an 8 cm×8 cm experimental prototype
1.A. D. Maidment and M. J. Yaffe, “Analysis of the spatial-frequency dependent DQE of optically coupled digital mammography detectors,” Med. Phys. 21, 721–729 (1994).
2.M. B. Williams, P. U. Simoni, L. Smilowitz, M. Stanton, W. Phillips, and A. Stewart, “Analysis of the detective quantum efficiency of a developmental detector for digital mammography,” Med. Phys. 26, 2273–2285 (1999).
3.S. Vedantham, A. Karellas, S. Suryanarayanan, D. Albagli, S. Han, E. J. Tkaczyk, C. E. Landberg, B. Opsahl-Ong, P. R. Granfors, I. Levis, C. J. D’Orsi, and R. E. Hendrick, “Full breast digital mammography with an amorphous silicon-based flat panel detector: Physical characteristics of a clinical prototype,” Med. Phys. 27, 558–567 (2000).
4.S. Vedantham, A. Karellas, S. Suryanarayanan, I. Levis, M. Sayag, R. Kleehammer, R. Heidsieck, and C. J. D’Orsi, “Mammographic Imaging with a small-format CCD-based digital cassette: Physical characteristics of a clinical system,” Med. Phys. 27, 1832–1840 (2000).
5.U. Neitzel, I. Maack, and S. Gunther-Kohfahl, “Image quality of a digital chest radiography system based on a selenium detector,” Med. Phys. 21, 509–516 (1994).
6.J. T. Dobbins, D. L. Ergun, L. Rutz, D. A. Hinshaw, H. Blume, and D. C. Clark, “DQE(f) of four generations of computed radiography acquisition devices,” Med. Phys. 22, 1581–1593 (1995).
7.W. Zhao, I. Blevis, S. Germann, J. A. Rowlands, D. Waechter, and Z. Huang, “Digital radiology using active matrix readout of amorphous selenium: construction and evaluation of a prototype real-time detector,” Med. Phys. 24, 1834–1843 (1997).
8.S. M. Kengyelics, J. H. Launders, and A. R. Cowen, “Physical imaging performance of a compact computed radiography acquisition device,” Med. Phys. 25, 354–360 (1998).
9.J. H. Siewerdsen, L. E. Antonuk, Y. el-Mohri, J. Yorkston, W. Huang, and I. A. Cunningham, “Signal, noise power spectrum, and detective quantum efficiency of indirect-detection flat-panel imagers for diagnostic radiology,” Med. Phys. 25, 614–628 (1998).
10.P. R. Granfors and R. Aufrichtig, “Performance of a 41×41-cm2 amorphous silicon flat panel x-ray detector for digital chest radiography,” Med. Phys. 27, 1324–1331 (2000).
11.C. E. Floyd, R. J. Warp, J. T. Dobbins, H. G. Chotas, A. H. Baydush, R. Vargas-Voracek, and C. E. Ravin, “Imaging characteristics of an amorphous silicon flat-panel detector for chest radiography,” Radiology 218, 683–688 (2001).
12.C. C. Shaw, X. Liu, S. Chardon, and J. M. Herron, “Frequency-dependent DQE in dual-screen CR imaging,” Proc. SPIE 3336, 684–694 (1998).
13.J. A. Rowlands and K. W. Taylor, “Detective quantum efficiency of x-ray image intensifiers: comparison of scintillation spectrum and rms methods,” Med. Phys. 11, 597–601 (1984).
14.G. Spekowius, H. Boerner, W. Eckenbach, and P. Quadfleig, “Simulation of the imaging performance of x-ray image intensifier/TV camera chains,” Proc. SPIE 2432, 12–23 (1995).
15.E. L. Baker, A. R. Cowen, R. Kemner, and R. Bastiaens, “Physical image quality evaluation of a CCD-based x-ray image intensifier digital fluorography system for cardiac applications,” Proc. SPIE 3336, 430–441 (1998).
16.N. W. Marshall, “The practical application of signal detection theory to image quality assessment in x-ray image intensifier-TV fluoroscopy,” Phys. Med. Biol. 46, 1631–1649 (2001).
17.H. Onihashi, H. Aida, K. Nou, T. Noji, Y. Murakoshi, K. Ito, K. Saito, and J. Koma, “Flat x-ray image intensifier system for real-time fluoroscopy,” Proc. SPIE 4320, 509–515 (2001).
18.P. R. Granfors, “Performance characteristics of an amorphous silicon flat panel x-ray imaging detector,” Proc. SPIE 3659, 480–490 (1999).
19.P. R. Granfors, D. Albagli, E. J. Tkaczyk, R. Aufrichtig, H. Netel, G. Brunst, J. M. Boudry, and D. Luo, “Performance of a flat-panel cardiac detector,” Proc. SPIE 4320, 77–86 (2001).
20.T. Bruijns, R. Bastiaens, B. Hoornaert, E. von Reth, F. Busse, V. Heer, T. Ducourant, A. Cowen, A. Davies, and F. Terrier, “Image quality of a large area dynamic flat detector: comparison with a state-of-the-art II/TV system,” Proc. SPIE 4682, 332–343 (2002).
21.F. Busse, W. Rutten, B. Sandkamp, P. L. Alving, R. J. M. Bastiaens, and T. Ducourant, “Design and performance of a high-quality cardiac flat detector,” Proc. SPIE 4682, 819–827 (2002).
22.S. Ikeda, K. Suzuki, K. Ishikawa, R. E. Colbeth, C. Webb, S. Tanaka, and K. Okusako, “Development and evaluation of a digital radiography system using a large-area flat panel detector,” Proc. SPIE 5030, 215–225 (2003).
23.S. Adachi, N. Hori, K. Sato, S. Tokuda, T. Sato, K. Uehara, Y. Izumi, H. Nagata, Y. Yoshimura, and S. Yamada, “Experimental evaluation of a-Se and CdTe flat panel x-ray detectors for digital radiography and fluoroscopy,” Proc. SPIE 3977, 38–47 (2000).
24.S. Yamada, H. Umazaki, A. Takahashi, M. Honda, K. Shiraishi, S. Rudin, D. R. Bednarek, C. J. Yang, Z. Wang, and A. Gopal, “Image quality evaluation of a selenium-based flat-panel digital x-ray detector system based on animal studies,” Proc. SPIE 3977, 429–436 (2000).
25.D. C. Hunt, O. Tousignant, Y. Demers, L. Laperriere, and J. A. Rowlands, “Imaging performance of amorphous selenium flat-panel detector for digital fluoroscopy,” Proc. SPIE 5030, 226–234 (2003).
26.P. Massoumzadeh, S. Rudin, and D. R. Bednarek, “Quantitative evaluation of amorphous selenium (a-Se) for region-of-interest fluoroscopy,” Proc. SPIE 3977, 658–669 (2000).
27.A. Ganguly, S. Rudin, D. R. Bednarek, K. R. Hoffman, and I. S. Kyprianou, “Microangiography for neurovascular imaging. I. Experimental evaluation and feasibility,” Med. Phys. 30, 3018–3028 (2003).
28.A. Ganguly, S. Rudin, D. R. Bednarek, and K. R. Hoffman, “Micro-angiography for neurovascular imaging. II. Cascade model analysis,” Med. Phys. 30, 3029–3039 (2003).
29.M. J. Flynn and E. Samei, “Experimental comparison of noise and resolution for and storage phosphor radiography systems,” Med. Phys. 26, 1612–1623 (1999).
30.K. Stierstorfer and M. Spahn, “Self-normalizing method to measure the detective quantum efficiency of a wide range of x-ray detectors,” Med. Phys. 26, 1312–1319 (1999).
31.E. Samei and M. J. Flynn, “An experimental comparison of detector performance for computed radiography systems,” Med. Phys. 29, 447–459 (2002).
32.E. Samei and M. J. Flynn, “An experimental comparison of detector performance for direct and indirect digital radiography systems,” Med. Phys. 30, 608–622 (2003).
33.E. Samei, “Image quality in two phosphor-based flat panel digital radiographic detector,” Med. Phys. 30, 1747–1757 (2003).
34.P. R. Granfors, R. Aufrichtig, G. E. Possin, B. W. Giambattista, Z. S. Huang, J. Liu, and B. Ma, “Performance of a 41×41 cm2 amorphous silicon flat panel x-ray detector designed for angiographic and R&F imaging applications,” Med. Phys. 30, 2715–2726 (2003).
35.The Institute of Physics and Engineering in Medicine (IPEM), Catalogue of Diagnostic X-ray Spectra and other Data, CD-ROM, Report No. 78 (The Institute of Physics and Engineering in Medicine, United Kingdom, 1997).
36.H. E. Johns and J. R. Cunningham, The Physics of Radiology, 4th ed. (C. C. Thomas, Springfield, 1983).
37.J. H. Siewerdsen, L. E. Antonuk, Y. El-Mohri, J. Yorkston, W. Huang, J. M. Boudry, and I. A. Cunningham, “Empirical and theoretical investigation of the noise performance of indirect detection, active matrix flat-panel imagers (AMFPIs) for diagnostic radiology,” Med. Phys. 24, 71–89 (1997).
38.M. Mahesh, “Fluoroscopy: patient radiation exposure issues,” Radiographics 21, 1033–1045 (2001).
39.J. M. Boone, D. E. Pfeiffer, K. J. Strauss, R. P. Rossi, P. J. Lin, J. S. Shepard, and B. J. Conway, “A survey of fluoroscopic exposure rates: AAPM Task Group No. 11 report,” Med. Phys. 20, 789–794 (1993).
40.D. P. Chakraborty, “Routine fluoroscopic quality control,” in Medical Physics Monograph No. 20: Specification, Acceptance Testing and Quality Control of Diagnostic X-ray Imaging Equipment, edited by J. A. Seibert, G. T. Barnes, and R. G. Gould (AIP, Woodbury, NY, 1994), pp. 569–595.
41.P.-J. P. Lin, “Technical considerations of equipment selection and acceptance testing of cardiovascular angiography systems,” in Medical Physics Monograph No. 20: Specification, Acceptance Testing and Quality Control of Diagnostic X-ray Imaging Equipment, edited by J. A. Seibert, G. T. Barnes, and R. G. Gould (AIP, Woodbury, NY, 1994), pp. 597–635.
42.H. Fujita, D. Y. Tsai, T. Itoh, K. Doi, J. Morishita, K. Ueda, and A. Ohtsuka, “A simple method for determining the modulation transfer function in digital radiography,” IEEE Trans. Med. Imaging MI-11, 34–39 (1992).
43.M. B. Williams, P. A. Mangiafico, and P. U. Simoni, “Noise power spectra of images from digital mammography detectors,” Med. Phys. 26, 1279–1293 (1999).
44.P. R. Granfors and R. Aufrichtig, “DQE(f) of an amorphous silicon flat-panel x-ray detector: detector parameter influences and measurement methodology,” Proc. SPIE 3977, 2–13 (2000).
45.R. Luhta and J. A. Rowlands, “Feasibility of large area x-ray sensitive vidicon for medical fluoroscopy: Resolution and lag factors,” Med. Phys. 24, 621–631 (1997).
46.J. A. Rowlands, “Television camera design and specification,” in Medical Physics Monograph No. 20: Specification, Acceptance Testing and Quality Control of Diagnostic X-ray Imaging Equipment, edited by J. A. Seibert, G. T. Barnes, and R. G. Gould (AIP, Woodbury, NY, 1994), pp. 461–481.
47.N. W. Marshall and C. J. Kotre, “Measurement and correction of the effects of lag on contrast-detail test results in fluoroscopy,” Phys. Med. Biol. 47, 947–960 (2002).
48.L. E. Antonuk, Y. El-Mohri, J. H. Siewerdsen, J. Yorkston, W. Huang, V. E. Scarpine, and R. A. Street, “Empirical investigation of the signal performance of a high-resolution, indirect-detection, active-matrix flat-panel imager (AMFPI) for fluoroscopic and radiographic operation,” Med. Phys. 24, 51–70 (1997).
49.O. Tousignant, Y. Demers, L. Laperriere, M. Nishiki, S. Nagai, T. Tomisaki, A. Takahashi, and K. Aoki, “Clinical performance of a 14″×14″ real time amorphous selenium flat panel detector,” Proc. SPIE 5030, 71–76 (2003).
50.M. Choquette, H. Rougeot, J.-P. Martin, L. Laperriere, Z. Shukri, and B. Polischuk, “Direct-selenium x-ray detector for fluoroscopy, R&F, and radiography,” Proc. SPIE 3977, 128–136 (2000).
51.J. H. Siewerdsen and D. A. Jaffray, “A ghost-story: spatio-temporal response characteristics of an indirect-detection flat-panel imager,” Med. Phys. 26, 1624–1641 (1999).
52.S. Vedantham, A. Karellas, and S. Suryanarayanan, “Solid-state fluoroscopic imager for high-resolution angiography: Parallel-cascaded linear systems analysis,” Med. Phys. 31, 1258–1268 (2004).
53.S. Rudin, Y. Wu, I. S. Kyprianou, C. N. Ionita, Z. Wang, A. Ganguly, and D. R. Bednarek, “Micro-angiographic detector with fluoroscopic capability,” Proc. SPIE 4682, 344–354 (2002).
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
Article metrics loading...