Full text loading...
No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
Response to the comments by Verellen, Soete, Erbel, and Labsik on the question of how to make rigid-body setup corrections in radiotherapy
1.M. J. Murphy, “Image-guided patient positioning: If one cannot correct for rotational errors in external-beam radiotherapy setup, should the rotations be measured?,” Med. Phys. 34, 1880–1883 (2007).http://dx.doi.org/10.1118/1.2731485
2.G. Soete, D. Verellen, K. Tournel, and G. Storme, “Setup accuracy of stereoscopic x-ray positioning with automated correction for rotational errors in patients treated with conformal arc radiotherapy for prostate cancer,” Radiother. Oncol. 80, 371–373 (2006).http://dx.doi.org/10.1016/j.radonc.2006.07.001
3.M. J. Murphy, “A six-degree-of-freedom image registration algorithm for image-guided frameless stereotaxic radiosurgery,” Med. Phys. 24, 857–866 (1997).http://dx.doi.org/10.1118/1.598005
4.M. J. Murphy, “The importance of CT slice thickness in Radiographic Patient Positioning for Radiosurgery,” Med. Phys. 26, 171–175 (1999).http://dx.doi.org/10.1118/1.598500
5.M. J. Murphy, “Fiducial-based targeting accuracy for external-beam radiotherapy,” Med. Phys. 29, 334–344 (2002).http://dx.doi.org/10.1118/1.1448823
Article metrics loading...