No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
A dynamic collimation system for penumbra reduction in spot-scanning proton therapy: Proof of concept
2. B. Arjomandy, N. Sahoo, X. R. Zhu, J. R. Zullo, R. Y. Wu, M. Zhu, X. Ding, C. Martin, G. Ciangaru, and M. T. Gillin, “An overview of the comprehensive proton therapy machine quality assurance procedures implemented at The University of Texas M. D. Anderson Cancer Center Proton Therapy Center-Houston,” Med. Phys. 36, 2269–2282 (2009).
3. J. B. Farr, A. E. Mascia, W. C. Hsi, C. E. Allgower, F. Jesseph, A. N. Schreuder, M. Wolanski, D. F. Nichiporov, and V. Anferov, “Clinical characterization of a proton beam continuous uniform scanning system with dose layer stacking,” Med. Phys. 35, 4945–4954 (2008).
4. S. Safai, T. Bortfeld, and M. Engelsman, “Comparison between the lateral penumbra of a collimated double-scattered beam and uncollimated scanning beam in proton radiotherapy,” Phys. Med. Biol. 53, 1729–1750 (2008).
5. A. J. Lomax, T. Bohringer, A. Bolsi, D. Coray, F. Emert, G. Goitein, M. Jermann, S. Lin, E. Pedroni, H. Rutz, O. Stadelmann, B. Timmermann, J. Verwey, and D. C. Weber, “Treatment planning and verification of proton therapy using spot scanning: initial experiences,” Med. Phys. 31, 3150–3157 (2004).
6. T. A. van de Water, A. J. Lomax, H. P. Bijl, C. Schilstra, E. B. Hug, and J. A. Langendijk, “Using a reduced spot size for intensity-modulated proton therapy potentially improves salivary gland-sparing in oropharyngeal cancer,” Int. J. Radiat. Oncol., Biol., Phys. 82, e313–e319 (2012).
7. J. Daartz, M. Bangert, M. R. Bussiere, M. Engelsman, and H. M. Kooy, “Characterization of a mini-multileaf collimator in a proton beamline,” Med. Phys. 36, 1886–1894 (2009).
8. M. Bues, W. D. Newhauser, U. Titt, and A. R. Smith, “Therapeutic step and shoot proton beam spot-scanning with a multi-leaf collimator: A Monte Carlo study,” Radiat. Prot. Dosim. 115, 164–169 (2005).
9. M. Torikoshi, S. Minohara, N. Kanematsu, M. Komori, M. Kanazawa, K. Noda, N. Miyahara, H. Itoh, M. Endo, and T. Kanai, “Irradiation system for HIMAC,” J. Radiat. Res. 48, A15–A25 (2007).
10. D. J. Brenner, C. D. Elliston, E. J. Hall, and H. Paganetti, “Reduction of the secondary neutron dose in passively scattered proton radiotherapy, using an optimized pre-collimator/collimator,” Phys. Med. Biol. 54, 6065–6078 (2009).
11. B. Dirksen, D. Wang, D. Hyer, J. Buatti, A. Sheybani, M. TenNapel, J. Bayouth, and R. Flynn, “Radiosurgery of peripheral lesions by spot scanning proton therapy,” Med. Phys. 40(6), 364 (2013).
Article metrics loading...
In the absence of a collimation system the lateral penumbra of spot scanning (SS) dose distributions delivered by low energy proton beams is highly dependent on the spot size. For current commercial equipment, spot size increases with decreasing proton energy thereby reducing the benefit of the SS technique. This paper presents a dynamic collimation system (DCS) for sharpening the lateral penumbra of proton therapy dose distributions delivered by SS.
The collimation system presented here exploits the property that a proton pencil beam used for SS requires collimation only when it is near the target edge, enabling the use of trimmers that are in motion at times when the pencil beam is away from the target edge. The device consists of two pairs of parallel nickel trimmer blades of 2 cm thickness and dimensions of 2 cm × 18 cm in the beam's eye view. The two pairs of trimmer blades are rotated 90° relative to each other to form a rectangular shape. Each trimmer blade is capable of rapid motion in the direction perpendicular to the central beam axis by means of a linear motor, with maximum velocity and acceleration of 2.5 m/s and 19.6 m/s2, respectively. The blades travel on curved tracks to match the divergence of the proton source. An algorithm for selecting blade positions is developed to minimize the dose delivered outside of the target, and treatment plans are created both with and without the DCS.
The snout of the DCS has outer dimensions of 22.6 × 22.6 cm2 and is capable of delivering a minimum treatment field size of 15 × 15 cm2. Using currently available components, the constructed system would weigh less than 20 kg. For irregularly shaped fields, the use of the DCS reduces the mean dose outside of a 2D target of 46.6 cm2 by approximately 40% as compared to an identical plan without collimation. The use of the DCS increased treatment time by 1–3 s per energy layer.
The spread of the lateral penumbra of low-energy SS proton treatments may be greatly reduced with the use of this system at the cost of only a small penalty in delivery time.
Full text loading...
Most read this month