No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
How does c
image quality compare with conventional 2D FFDM?
1.T. M. Svahn, D. P. Chakraborty, D. Ikeda, S. Zackrisson, Y. Do, S. Mattsson, and I. Andersson, “Breast tomosynthesis and digital mammography: A comparison of diagnostic accuracy,” Br. J. Radiol. 85, e1074–e1082 (2012).
4.U.S. Food Drug Administration, P080003/S001 Hologic Selenia Dimensions 3D System, FDA Summary of Safety and Effectiveness Data, FDA,2013.
5.U.S. Food Drug Administration, P080003/S001 Hologic Selenia Dimensions 3D System, FDA Executive Summary, FDA, 2012.
6.M. L. Zuley, B. Guo, V. J. Catullo, D. M. Chough, A. E. Kelly, A. H. Lu, G. Y. Rathfon, M. Lee Spangler, J. H. Sumkin, L. P. Wallace, and A. I. Bandos, “Comparison of two-dimensional synthesized mammograms versus original digital mammograms alone and in combination with tomosynthesis images,” Radiology 271, 664–671 (2014).
7.P. Skaane, A. I. Bandos, E. B. Eben, I. N. Jebsen, M. Krager, U. Haakenaasen, U. Ekseth, M. Izadi, S. Hofvind, and R. Gullien, “Two-view digital breast tomosynthesis screening with synthetically reconstructed projection images: Comparison with digital breast tomosynthesis with full-field digital mammographic images,” Radiology 271, 655–663 (2014).
8.N. Kiarashi, A. C. Nolte, G. M. Sturgeon, W. P. Segars, S. V. Ghate, L. W. Nolte, E. Samei, and J. Y. Lo, “Development of realistic physical breast phantoms matched to virtual breast phantoms based on human subject data,” Med. Phys. 42, 4116–4126 (2015).
9.R. Hendrick, L. Bassett, M. Botsco, D. Deibel, S. Feig, J. Gray, A. Haus, R. Heinlein, E. Kitts, and J. McCrohan, Mammography Quality Control Manual (American College of Radiology, Reston, Va, 1999).
10.K. W. Brooks, J. H. Trueblood, K. J. Kearfott, and D. T. Lawton, “Automated analysis of the American College of Radiology mammographic accreditation phantom images,” Med. Phys. 24, 709–723 (1997).
11.A. Rose, Vision: Human and Electronic (Plenum, New York, NY, 1974).
12.R. S. Saunders, Jr. and E. Samei, “A method for modifying the image quality parameters of digital radiographic images,” Med. Phys. 30, 3006–3017 (2003).
13.R. S. Saunders, Jr., J. A. Baker, D. M. Delong, J. P. Johnson, and E. Samei, “Does image quality matter? Impact of resolution and noise on mammographic task performance,” Med. Phys. 34, 3971–3981 (2007).
15.R. S. Saunders, E. Samei, and C. Hoeschen, “Impact of resolution and noise characteristics of digital radiographic detectors on the detectability of lung nodules,” Med. Phys. 31, 1603–1613 (2004).
16.U.S. Food Drug Administration P080003/S001 Hologic Selenia Dimensionsc-view Software Module, Sponsor Executive Summary, FDA, 2013.
17.A. E. Burgess, F. L. Jacobson, and P. F. Judy, “Human observer detection experiments with mammograms and power-law noise,” Med. Phys. 28, 419–437 (2001).
18.J. M. Park, E. A. Franken, Jr., M. Garg, L. L. Fajardo, and L. T. Niklason, “Breast Tomosynthesis: Present considerations and future applications,” Radiographics 27, S231–S240 (2007).
Article metrics loading...
The FDA approved the use of digital breast tomosynthesis (DBT) in 2011 as an adjunct to 2D full field digital mammography (FFDM) with the constraint that all DBT acquisitions must be paired with a 2D image to assure adequate interpretative information is provided. Recently manufacturers have developed methods to provide a synthesized 2D image generated from the DBT data with the hope of sparing patients the radiation exposure from the FFDM acquisition. While this much needed alternative effectively reduces the total radiation burden, differences in image quality must also be considered. The goal of this study was to compare the intrinsic image quality of synthesized 2D c-view and 2D FFDM images in terms of resolution, contrast, and noise.
Two phantoms were utilized in this study: the American College of Radiology mammography accreditation phantom (ACR phantom) and a novel 3D printed anthropomorphic breast phantom. Both phantoms were imaged using a Hologic Selenia Dimensions 3D system. Analysis of the ACR phantom includes both visual inspection and objective automated analysis using in-house software.Analysis of the 3D anthropomorphic phantom includes visual assessment of resolution and Fourier analysis of the noise.
Using ACR-defined scoring criteria for the ACR phantom, the FFDM images scored statistically higher than c-view according to both the average observer and automated scores. In addition, between 50% and 70% of c-viewimages failed to meet the nominal minimum ACR accreditation requirements—primarily due to fiber breaks. Softwareanalysis demonstrated that c-view provided enhanced visualization of medium and large microcalcification objects; however, the benefits diminished for smaller high contrast objects and all low contrast objects. Visual analysis of the anthropomorphic phantom showed a measureable loss of resolution in the c-viewimage (11 lp/mm FFDM, 5 lp/mm c-view) and loss in detection of small microcalcification objects. Spectral analysis of the anthropomorphic phantom showed higher total noise magnitude in the FFDM image compared with c-view. Whereas the FFDM image contained approximately white noise texture, the c-viewimage exhibited marked noise reduction at midfrequency and high frequency with far less noise suppression at low frequencies resulting in a mottled noise appearance.
Their analysis demonstrates many instances where the c-viewimage quality differs from FFDM. Compared to FFDM, c-view offers a better depiction of objects of certain size and contrast, but provides poorer overall resolution and noise properties. Based on these findings, the utilization of c-viewimages in the clinical setting requires careful consideration, especially if considering the discontinuation of FFDM imaging. Not explicitly explored in this study is how the combination of DBT + c-view performs relative to DBT + FFDM or FFDM alone.
Full text loading...
Most read this month