Full text loading...
No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
Importance of the indentation depth in tapping-mode atomic force microscopy study of compliant materials
1.Q. Zhong, D. Innis, K. Kjoller, and V. B. Elings, Surf. Sci. Lett. 290, L688 (1993).
2.D. A. Chernoff, Proceedings Microscopy and Microanalysis (Jones and Begell, New York, 1995).
3.For some recent reviews, see O. I. Kiselyova and I. V. Yaminsky, Colloid J. USSR 61, 1 (1999);
3.K. D. Jandt, Mater. Sci. Eng. 21, 221 (1998);
3.S. N. Magonov and D. H. Reneker, Annu. Rev. Mater. Sci. 27, 175 (1997);
3.V. V. Tsukruk, Rubber Chem. Technol. 70, 430 (1997).
4.J. Tamayo and R. Garcia, Appl. Phys. Lett. 71, 2394 (1997).
5.M.-H. Whangbo, R. Brandsch, and G. Bar, Surf. Sci. 411, L794 (1998).
6.G. Bar, R. Brandsch, and M.-H. Whangbo, Surf. Sci. 411, L802 (1998).
7.O. P. Behrend, L. Odoni, J. L. Loubet, and N. A. Burnham (unpublished).
8.M. Marth, D. Maier, J. Honerkamp, R. Brandsch, and G. Bar, J. Appl. Phys. 85, 7030 (1999).
9.The measured raw data are the root mean square (rms) values of the amplitude which are related to the amplitude by a factor of √2. To calculate δ, it is necessary to calibrate the amplitude A. This can be achieved from the slope of curves on stiff samples (e.g., mica, Si). The quality factor of the oscillation can be obtained from the width at the half-power point of the amplitude vs frequency curve.
10.S. N. Magonov, V. Elings, and M.-H. Whangbo, Surf. Sci. 375, L385 (1997).
Article metrics loading...