No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
Comment on “A versatile thermoelectric temperature controller with 10 mK reproducibility and 100 mK absolute accuracy” [Rev. Sci. Instrum.80, 126107 (2009)]
Article metrics loading...
Libbrecht and Libbrecht recently described a thermoelectric temperature controller for which they claimed an absolute accuracy of 100 mK. They did not specify the heat-dissipation coefficient for their thermistor temperature sensor—which would dissipate more power than is usual in such circuits— nor make any allowance for self-heating in the thermistor. Self-heating can be expected to have introduced more than 100 mK of absolute error in their circuit.
Full text loading...
Most read this month