No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
A 10 mK scanning tunneling microscope operating in ultra high vacuum and high magnetic fields
8. C. J. Chen, Introduction to Scanning Tunneling Microscopy (Oxford University Press, 1993).
19. Y. J. Song, A. F. Otte, V. Shvarts, Z. Zhao, Y. Kuk, S. R. Blankenship, A. Band, F. M. Hess, and J. A. Stroscio, Rev. Sci. Instrum. 81, 121101 (2010).
20. M. Assig, “Development of a Millikelvin scanning tunneling microscope for applications in ultra high vacuum and high magnetic fields (No 5204),” Ph.D. dissertation, École Polytechnique Fédérale de Lausanne, 2011.
22. attocube systems AG. ANPz50/LT/UHV z-positioner.
23. H. London, in Proceedings of International Conference on Low Temperature Physics (International Union of Pure and Applied Physics, Oxford, UK, 1951), p. 157.
25. T. P. Das, R. de Bruyn Ouboter, and K. W. Taconis, in Proceedings of 9th International Conference on Low Temperature Physics (Plenum Press, 1965), p. 1253.
26. O. V. Lounasmaa, Experimental Principles and Methods Below 1K (Academic Press Inc., 1974).
27. F. Pobell, Matter and Methods at Low Temperatures, 3rd ed. (Springer, 2007).
29. J. C. Wheatley, O. E. Vilches, and W. R. Abel, Phys. 4, 1 (1968).
38.The inductance is measured with an Agilent 4263B LCR meter. The excitation voltage is 100 mV at a lock-in frequency of 120 Hz.
40.The resistances are measured with the Lakeshore 370S resistance bridge. The excitation voltages range from 632 μV at the 1 K-pot to 20 μV at the mixing chamber. The lock-in frequency is 9.8 Hz.
41. S. N. Mott, Conduction in Non-Crystalline Materials (Oxford University Press, 1993).
43. Spectrum control Inc. 51-726-017.
44. Spectrum control Inc. 1289-004.
Article metrics loading...
We present design and performance of a scanning tunneling microscope (STM) that operates at temperatures down to 10 mK providing ultimate energy resolution on the atomic scale. The STM is attached to a dilution refrigerator with direct access to an ultra high vacuum chamber allowing in situ sample preparation. High magnetic fields of up to 14 T perpendicular and up to 0.5 T parallel to the sample surface can be applied. Temperature sensors mounted directly at the tip and sample position verified the base temperature within a small error margin. Using a superconducting Al tip and a metallic Cu(111) sample, we determined an effective temperature of 38 ± 1 mK from the thermal broadening observed in the tunneling spectra. This results in an upper limit for the energy resolution of ΔE = 3.5k B T = 11.4 ± 0.3 μeV. The stability between tip and sample is 4 pm at a temperature of 15 mK as demonstrated by topography measurements on a Cu(111) surface.
Full text loading...
Most read this month