No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
Note: Magnetic noise from the inner wall of a magnetically shielded room
3. D. Cohen, U. Schlapfer, S. Ahlfors, M. Hamalainen, and E. Halgren, in Biomag 96: Proceedings of the 13th International Conference on Biomagnetism, edited by H. Nowak, J. Haueisen, F. Giessler, and R. Huonker (VDE, Berlin, 1997), p. 399.
5. J. Nenonen and T. Katila, in: Biomag 87: Proceedings of the 6th International Conference on Biomagnetism, edited by K. Atsumi, M. Kotani, S. Ueno, T. Katila, and S. J. Williamson (Denki University Press, Tokyo, 1988), p. 426.
Article metrics loading...
We measured the thermal magnetic noise generated by the inner high-permeability wall of a magnetically shielded room. This room houses a magnetoencephalogram (MEG), which contains 102 “small” identical magnetometers. For the measurement, we created two large magnetometers by summing the outputs of 46 magnetometers equally on the helmet's left and right side, to look at the summed noise of the right and left vertical walls. From these summed outputs, we calculated the rms noise amplitude due to all six walls at the MEG location to be ∼0.5 f T/ at 100 Hz, only slowly rising with lower frequency. This is well below the system noise of each small MEG magnetometer, hence is negligible for the MEG.
Full text loading...
Most read this month