No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
Wide-area scanner for high-speed atomic force microscopy
1. M. B. Viani, L. I. Pietrasanta, J. B. Thompson, A. Chand, I. C. Gebeshuber, J. H. Kindt, M. Richter, H. G. Hansma, and P. K. Hansma, Nat. Struct. Biol. 7, 644 (2000).
13. K. Igarashi, T. Uchihashi, A. Koivula, M. Wada, S. Kimura, T. Okamoto, M. Penttilä, T. Ando, and M. Samejima, Science 333, 1279 (2011).
27. A. J. Fleming and A. G. Wills, in Proceedings of the IFAC World Congress, Seoul, Korea, 7–11 July (Elsevier, Netherlands, 2008), p. 11805.
36. A. Ryter and O. E. Landman, J. Bacteriol. 88, 457 (1964).
37. J. Ghuysen, Bacteriol. Rev. 32, 425 (1968).
38. T. J. Smith, S. A. Blackman, and S. J. Foster, Microbiology 146, 249 (2000).
42. M. A. Barbieri, G. Li, M. I. Colombo, and P. D. Stahl, J. Biol. Chem. 269, 18720–18722 (1994).
44. A. I. Shevchuk, P. Novak, M. Taylor, I. A. Diakonov, A. Ziyadeh-Isleem, M. Bitoun, P. Guicheney, M. J. Lab, J. Gorelik, C. J. Merrifield, D. Klenerman, and Y. E. Korchev, J. Cell. Biol. 197, 499–508 (2012).
Article metrics loading...
High-speed atomic force microscopy (HS-AFM) has recently been established. The dynamic processes and structural dynamics of protein molecules in action have been successfully visualized using HS-AFM. However, its maximum scan ranges in the X- and Y-directions have been limited to ∼1 μm and ∼4 μm, respectively, making it infeasible to observe the dynamics of much larger samples, including live cells. Here, we develop a wide-area scanner with a maximum XY scan range of ∼46 × 46 μm2 by magnifying the displacements of stack piezoelectric actuators using a leverage mechanism. Mechanical vibrations produced by fast displacement of the X-scanner are suppressed by a combination of feed-forward inverse compensation and the use of triangular scan signals with rounded vertices. As a result, the scan speed in the X-direction reaches 6.3 mm/s even for a scan size as large as ∼40 μm. The nonlinearity of the X- and Y-piezoelectric actuators’ displacements that arises from their hysteresis is eliminated by polynomial-approximation-based open-loop control. The interference between the X- and Y-scanners is also eliminated by the same technique. The usefulness of this wide-area scanner is demonstrated by video imaging of dynamic processes in live bacterial and eukaryotic cells.
Full text loading...
Most read this month