No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
A low-noise transimpedance amplifier for the detection of “Violin-Mode” resonances in advanced Laser Interferometer Gravitational wave Observatory suspensions
5. A. Heptonstall et al., “Invited Article: CO2 laser production of fused silica fibres for use in interferometric gravitational wave detector mirror suspensions,” Rev. Sci. Instrum. 82, 011301–1011301–9 (2011).
7. N. A. Lockerbie, K. V. Tokmakov, and K. A. Strain, “A source of illumination for low-noise ‘Violin Mode’ shadow sensors, intended for use in interferometric gravitational wave detectors,” Meas. Sci. Technol. (to be published).
9. N. A. Lockerbie, L. Carbone, B. Shapiro, K. V. Tokmakov, A. Bell, and K. A. Strain “First results from the ‘Violin-Mode’ tests on an advanced LIGO suspension at MIT,”Classical Quantum Gravity 28, 245001 (2011).
11. A. V. Dmitriev, S. D. Mescheriakov,K. V. Tokmakov, and V. P. Mitrofanov, “Controllable damping of high-Q violin modes in fused silica suspension fibers,” Classical Quantum Gravity 27, 025009 (2010).
13. M. Stitt and W. Meinel, “OPT201 Photodiode-Amplifier Rejects Ambient Light,” Burr-Brown IC Applications Handbook (1993), p. 379.
14. M. Johnson, Photodetection and Measurement (McGraw-Hill, 2003), p. 146.
16. N. A. Lockerbie and K. V. Tokmakov, “Quasi-static displacement calibration system for a ‘Violin-Mode’ shadow-sensor intended for gravitational wave detector suspensions,” Rev. Sci. Instrum. 85, 105003 (2014).
17. N. A. Lockerbie
and K. V. Tokmakov
, A modulated Near InfraRed gain calibration system for a ‘Violin-Mode’ transimpedance amplifier, intended for advanced LIGO suspensions (TBA) LIGO-G1401275-v1, available at https://dcc.ligo.org
Article metrics loading...
This paper describes the design and performance of an extremely low-noise differential transimpedance amplifier, which takes its two inputs from separate photodiodes. The amplifier was planned to serve as the front-end electronics for a highly sensitive shadow-displacement sensing system, aimed at detecting very low-level “Violin-Mode” (VM) oscillations in 0.4 mm diameter by 600 mm long fused-silica suspension fibres. Four such highly tensioned fibres support the 40 kg test-masses/mirrors of the Advanced Laser Interferometer Gravitational wave Observatory interferometers. This novel design of amplifier incorporates features which prevent “noise-gain peaking” arising from large area photodiode (and cable) capacitances, and which also usefully separate the DC and AC photocurrents coming from the photodiodes. In consequence, the differential amplifier was able to generate straightforwardly two DC outputs, one per photodiode, as well as a single high-gain output for monitoring the VM oscillations—this output being derived from the difference of the photodiodes’ two, naturally anti-phase, AC photocurrents. Following a displacement calibration, the amplifier's final VM signal output was found to have an AC displacement responsivity at 500 Hz of (9.43 ± 1.20) MV(rms) m−1(rms), and, therefore, a shot-noise limited sensitivity to such AC shadow- (i.e., fibre-) displacements of (69 ± 13) picometres/√Hz at this frequency, over a measuring span of ±0.1 mm.
Full text loading...
Most read this month