No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
Laboratory-size three-dimensional x-ray microscope with Wolter type I mirror optics and an electron-impact water window x-ray source
3. D. Rudolph, B. Niemann, G. Schmahl, and O. Christ, X-Ray Microscopy, Springer Series in Optical Sciences Vol. 43, edited by G. Schmahl, and D. Rudolph (Springer, Berlin, Heidelberg, 1984), pp. 192–202.
4. W. Meyer-Ilse, H. Medecki, L. Jochum, E. Anderson, D. Attwood, C. Magowan, R. Balhorn, M. Moronne, D. Rudolph, and G. Schmahl, Synchrotron Radiat. News 8, 29 (1995).
6. G. Schneider, P. Guttmann, S. Heim, S. Rehbein, F. Mueller, K. Nagashima, J. B. Heymann, W. G. Muller, and J. G. McNally, Nat. Methods 7, 985 (2010).
11. H. Legall, G. Blobel, H. Stiel, W. Sandner, C. Seim, P. Takman, D. H. Martz, M. Selin, U. Vogt, H. M. Hertz, D. Esser, H. Sipma, J. Luttmann, M. Höfer, H. D. Hoffmann, S. Yulin, T. Feigl, S. Rehbein, P. Guttmann, G. Schneider, U. Wiesemann, M. Wirtz, and W. Diete, Opt. Express 20, 18362 (2012).
13. S. Ohsuka, A. Ohba, M. Sugiyama, T. Hayakawa, T. Matsumura, K. Kinoshita, N. Watanabe, Y. Shimanuki, Y. Sano, and H. Kihara, X-Ray Microscopy III, Springer Series in Optical Sciences Vol. 67, edited by A. Michette, G. Morrison, and C. Buckley (Springer, Berlin, Heidelberg, 1992), pp. 164–166.
17. A. Ohba, T. Nakano, S. Onoda, and M. Sugiyama, in Proceedings of the 8th International Conference on X-ray Microscopy (Institute of Pure and Applied Physics, 2006), pp. 50–52.
Article metrics loading...
We constructed a laboratory-size three-dimensional water window x-ray microscope that combines wide-field transmission x-ray microscopy with tomographic reconstruction techniques, and observed bio-medical samples to evaluate its applicability to life science research fields. It consists of a condenser and an objective grazing incidence Wolter type I mirror, an electron-impact type oxygen Kα x-ray source, and a back-illuminated CCD for x-ray imaging. A spatial resolution limit of around 1.0 line pairs per micrometer was obtained for two-dimensional transmission images, and 1-μm scale three-dimensional fine structures were resolved.
Full text loading...
Most read this month