- Conference date: 6–7 November 2009
- Location: Firenze (Italy)
Nd:YAG, currently used since 1970 in dental laboratories to weld metals on dental prostheses has some limits such great dimensions, high costs and fixed delivery system. Recently it was proposed the possibility to use the Nd:YAG laser device commonly utilised in dental office, to repair broken fixed, removable and orthodontic prostheses and to weld metals directly into the mouth.
The aim of this work is to value, through SEM (Scanning Electron Microscope), EDS (Energy Dispersive X‐Ray Spectroscopy) and DMA (Dynamic Mechanical Analysis), quality and mechanical strength of the welding process comparing a device normally used in dental lab and a device normally used in dental office for oral surgery.
Sixteen CoCrMo metal plates and twenty steel orthodontic wires were divided in four groups: one was welded without metal apposition by laboratory laser, one was welded with metal apposition by laboratory laser, one was welded without metal apposition by office laser and one was welded with metal apposition by office laser. The welding process was analysed by SEM, EDS and DMA to compare the differences between the different samples.
By SEM analysis it was seen that the plates welded by office laser without apposition metal showed a greater number of fissurations compared with the other samples. By EDS analysis it was seen a homogeneous composition of the metals in all the samples. The mechanical tests showed a similar elastic behaviour of the samples, with minimal differences between the two devices. No wire broke even under the maximum strength by the Analyser.
This study seems to demonstrate that the welding process by office Nd:YAG laser device and the welding process by laboratory Nd:YAG laser device, analysed by SEM, EDS and DMA, showed minimal and not significant differences even if these data will be confirmed by a greater number of samples.
- Scanning electron microscopy
- Nd:YAG lasers
- Laser beam welding
- Dynamic mechanical analysis
- Electron spectroscopy
- Mechanical energy
- Mechanical testing
MOST READ THIS MONTH
MOST CITED THIS MONTH
Y. K. Semertzidis, M. Aoki, M. Auzinsh, V. Balakin, A. Bazhan, G. W. Bennett, R. M. Carey, P. Cushman, P. T. Debevec, A. Dudnikov, F. J. M. Farley, D. W. Hertzog, M. Iwasaki, K. Jungmann, D. Kawall, B. Khazin, I. B. Khriplovich, B. Kirk, Y. Kuno, D. M. Lazarus, L. B. Leipuner, V. Logashenko, K. R. Lynch, W. J. Marciano, R. McNabb, W. Meng, J. P. Miller, W. M. Morse, C. J. G. Onderwater, Y. F. Orlov, C. S. Ozben, R. Prigl, S. Rescia, B. L. Roberts, N. Shafer‐Ray, A. Silenko, E. J. Stephenson, K. Yoshimura and EDM Collaboration
Article metrics loading...