Monitoring the lesion formation during histotripsy treatment using shear wave imaging
- Conference date: 10–13 June 2012
- Location: Heidelberg, Germany
Monitoring the lesion formation induced by histotripsy has mainly relied on the quantitative change in backscatter intensity using ultrasound B-mode imaging. However, how the mechanical properties of the histotripsy-treated tissue region alter during the procedure is yet to be fully investigated. We thus proposed here to monitor such a therapeutic process based on shear modulus estimated by shear wave imaging (SWI). In the therapeutic procedure, a single-element piezo-composite focused transducer (Imasonic, Besançon, France) with a center frequency of 660 kHz, a focal length of 45 mm, and an fnumber of 1 was driven by a function generator (AFG 3101, Tektronix, Beaverton, OR) and a gated RF power amplifier (GA-2500A, RITEC Inc., USA) to generate ultrasound histotripsy pulses. Histotripsy pulses were delivered for 20 seconds and then followed by a 30-second pause and a rapid monitoring step. Such a treatment and monitoring scheme was repeated for 10 mins. Both the reference measurement and monitoring were realized by SWI, where plane shear waves were generated by an 8 MHz linear array probe connected to a prototype ultrasound scanner, and acquired at a frame rate of 10000 Hz. Shear modulus was estimated and mapped in 2D through a time-of-flight algorithm. Gelatin (8%)-agar (2%) phantoms and ex-vivo porcine liver samples were tested. Regions of interests (ROI's) of 2 mm-by-2 mm in both untreated and treated regions were selected to compute the contrast-to-noise ratio (CNR). In all three scenarios where different PD's and PRF's were implemented, during the first 100 seconds of the treatment, 50% decrease in the shear modulus within the histotripsy-targeted zone was already observed, and the CNR of the shear modulus increased by 18 dB. In contrast, the backscatter intensity began to reduce and the corresponding CNR was found to increase by 6 dB only after 120 seconds of treatment. The results demonstrated that SWI can map quantitatively the change of mechanical properties during histotripsy treatment. Moreover, the shear modulus estimated by SWI was a more sensitive indicator of the lesion formation than the backscatter intensity obtained from B-mode at the early stage of the histotripsy treatment. In-vitro experiments on liver samples have also been carried out.
- Elastic moduli
- Medical image noise
- Medical imaging
- Mechanical properties
- Signal generators
- Functional monitoring and imaging
MOST READ THIS MONTH
MOST CITED THIS MONTH
Y. K. Semertzidis, M. Aoki, M. Auzinsh, V. Balakin, A. Bazhan, G. W. Bennett, R. M. Carey, P. Cushman, P. T. Debevec, A. Dudnikov, F. J. M. Farley, D. W. Hertzog, M. Iwasaki, K. Jungmann, D. Kawall, B. Khazin, I. B. Khriplovich, B. Kirk, Y. Kuno, D. M. Lazarus, L. B. Leipuner, V. Logashenko, K. R. Lynch, W. J. Marciano, R. McNabb, W. Meng, J. P. Miller, W. M. Morse, C. J. G. Onderwater, Y. F. Orlov, C. S. Ozben, R. Prigl, S. Rescia, B. L. Roberts, N. Shafer‐Ray, A. Silenko, E. J. Stephenson, K. Yoshimura and EDM Collaboration
Article metrics loading...