No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
Speech perception in noise with a two-sensor frequency-domain minimum-variance (FMV) beamforming algorithm
1.Agnew, J. , and Thornton, J.M. (2000). “Just noticeable and objectionable group delays in digital hearing aids,” J. Am. Acad. Audio. 11, 330–336.
2.Bilger, R.C. , Neutzel, J.M. , and Rabinowitz, W.M. (1984). “Standardization of a test of speech perception in noise,” J. Speech Hear. Res. 21, 5–36.
3.Capon, J. (1969). “High-resolution frequency-wavenumber spectrum analysis,” Proc. IEEE 57(8), 1408–1419.
4.Cox, R.M. , and McDaniel, D.M. (1989). “Development of the Speech Intelligibility Rating (SIR) Test for hearing aid comparisons,” J. Speech Hear. Res. 32, 347–352.
5.Elledge, M.E. , Lockwood, M.E. , Bilger, R.C. , Feng, A.S. , Goueygou, M. , Jones, D.L. , Lansing, C.R. , Liu, C. , O’Brien, W.D. Jr., and Wheeler, B.C. (1999). “A real-time dual-microphone signal-processing system for hearing aids,” J. Acoust. Soc. Am. 106 (Pt. 2), 2279A.
6.Fischer, S. and Simmer, K.U. (1996). “Beamforming microphone arrays for speech acquisition in noisy environments,” Speech Comm. 20, 215–227.
7.Griffiths, L.J. , and Jim, C.W. (1982). “An alternative approach to linearly constrained adaptive beamforming,” IEEE Trans. Antennas Propag. AP-30(1), 27–34.
8.Larsen, J.B. , Lockwood, M.E. , Lansing, C.R. , Bilger, R.C. , Wheeler, B.C. , O’Brien, W.D. , Jones, D.L. , and Feng, A.S. (2001). “Human performance in a multisource environment with a frequency-banded minimum-variance beamforming algorithm,” J. Acoust. Soc. Am. 109(5), 2494.
9.Lockwood, M.E. , Jones, D. , Bilger, R.C. , Lansing, C.R. , O’Brien, W.D. , Wheeler, B.C. , and Feng, A.S. (2004). “Performance of time- and frequency-domain binaural beamformers based on recorded signals from real rooms,” J. Acoust. Soc. Am. 115(1), 379–391.
10.Nilsson, M.J. , Soli, S.D. , and Sullivan, J. (1994). “Development of a hearing in noise test for the measurement of speech reception threshold,” J. Acoust. Soc. Am. 95, 1985–1999.
11.Ricketts, T.A. and Dahr, S. (1999). “Aided benefit across directional and omni-directional hearing aid microphones for behind-the-ear hearing aids,” J. Am. Acad. Audio. 10, 180–189.
12.Weiner, B.J. (1962). Statistical Principles in Experimental Design (McGraw-Hill, New York), pp. 221–222.
13.Yang, K.L., Lockwood, M.E., Elledge, M.E., and Jones, D.L. (2000). “A comparison of beamforming algorithms for binaural acoustic processing,” Proc. 9th IEEE Digital Signal Processing Workshop, Hunt, TX, 15–18 October.
14.Zurek, P.M., Greenberg, J.E., and Rabinowitz, W.M. (1996). “Prospects and limitations of microphone-array hearing aids,” in Psychoacoustics, Speech and Hearing Aids, edited by B. Kollmeier (World Scientific, Singapore).
Article metrics loading...
Full text loading...