No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
Design and implementation of a shielded underwater vector sensor for laboratory environments
1. T. B. Gabrielson, D. L. Gardner, and S. L. Garrett, “A simple neutrally buoyant sensor for direct measurement of particle velocity and intensity in water,” J. Acoust. Soc. Am. 97(1), 2227–2237 (1995).
3. K. Kim, T. B. Gabrielson, and G. C. Lauchle, “Development of an accelerometer based underwater acoustic intensity sensor,” J. Acoust. Soc. Am. 116(6), 3384–3392 (2004).
4. K. J. Bastyr, G. C. Lauchle, and J. A. McConnell, “Development of a velocity gradient underwater acoustic intensity sensor,” J. Acoust. Soc. Am. 106(6), 3178–3188 (1999).
5. J. A. McConnell, “Development and application of inertial type underwater acoustic intensity probes,” Ph.D. dissertation, Penn State University, University Park, PA, 2004.
6. J. S. Bendat and A. G. Piersol, Random Data Analysis and Measurement Procedures, 3rd ed. (Wiley and Sons, New York, 2000).
Article metrics loading...
Underwater acoustic vector sensors, for measuring acoustic intensity, are typically used in open water where electromagnetic interference (EMI) is generally not a contributor to overall background noise. However, vector sensors are also useful in a laboratory setting where EMI can be a limiting factor at low frequencies. An underwater vector sensor is designed and built with specific care for EMI immunity. The sensor, and associated signal processing, is shown to reduce background noise at EMI frequencies by 10–50 dB and 10–20 dB across the entire frequency bandwidth, as compared to an identical unshielded vector sensor.
Full text loading...
Most read this month