No data available.
Please log in to see this content.
You have no subscription access to this content.
No metrics data to plot.
The attempt to load metrics for this article has failed.
The attempt to plot a graph for these metrics has failed.
The full text of this article is not currently available.
Evaluation of smartphone sound measurement applicationsa)
1. ANSI (1983). ANSI S1.4-1983, Specification for Sound Level Meters (American National Standards Institute, New York) [1983 (R2007)].
2.European Environment Agency. (2013). eyeonearth.org. http://eyeonearth.org/map/NoiseWatch/ (Last viewed June 23, 2013).
3. Kanhere, S. S. (2013). “ Participatory sensing: Crowdsourcing data from mobile smartphones in urban spaces,” in Distributed Computing and Internet Technology (Springer, Berlin), pp. 19–26.
5. Maisonneuve, N. , and Matthias, N. (2010). “ Participatory noise pollution monitoring using mobile phones” Inf. Polity 15, 51–71.
6. Maisonneuve, N. , Mattias, S. , Nissen, M. E. , Hanappe, P. , and Steels, L. (2009). “ Citizen noise pollution monitoring,” in Proceedings of the 10th Annual International Conference on Digital Government Research: Social Networks: Making Connections between Citizens, Data and Government, Digital Government Society of North America, pp. 96–103.
8. Sieber, W. K. (1991). “ Development, use, and availability of a job exposure matrix based on national occupational hazard survey data,” Am. J. Ind. Med. 20(2), 163–174.
Article metrics loading...
This study reports on the accuracy of smartphone sound measurement applications (apps) and whether they can be appropriately employed for occupational noise measurements. A representative sample of smartphones and tablets on various platforms were acquired, more than 130 iOS apps were evaluated but only 10 apps met our selection criteria. Only 4 out of 62 Android apps were tested. The results showed two apps with mean differences of 0.07 dB (unweighted) and −0.52 dB (A-weighted) from the reference values. Two other apps had mean differences within ±2 dB. The study suggests that certain apps may be appropriate for use in occupational noise measurements.
Full text loading...
Most read this month